These cookies are necessary for the basic functions of the shop.
"Allow all cookies" cookie
"Decline all cookies" cookie
Change gross and net price
These cookies are used to make the shopping experience even more appealing, for example for the recognition of the visitor.
Statistics & Tracking
Conversion and usertracking via Google Tag Manager
Track device being used
Combotester X3 with foot electrode
Prices incl. VAT plus shipping costs
Shoe- and wirst strap test station X3, with foot electrode and wall mount. The Combotester X3... more
Shoe- and wirst strap test station X3, with foot electrode and wall mount.
The Combotester X3 checks the functionality of a users wirst strap and shoes. Both shoes and the wrist strap will be checked against the threshold of 750 kΩ - 35 MΩ. Thresholds are selectable via DIP switch. Measuring results are displayed via status-LEDs. Through the potential-free relay contact an entrance system can be opened after a successful test.
The Combotester X3 comes with a 2-pole foot electrode and a wall mount.
- Thresholds selectable via DIP switch
- Status LEDs display measuring results
- Test modi selectable via DIP switch
- Relay outlet
- Combo X3 personnel grounding tester
- Power supply 12V DC
- Holding plate for wall mounting
- 2-pole foot electrode
- Calibration certificate
|Accuracy||± 20% for 1 GΩ shoe threshold |
± 10% for all other thresholds
|Operating voltage||12V DC|
|Voltage test switch||5V DC, opened|
|Voltage for wrist strap-and-shoe test||30V DC, opened|
|Testflows||Reliant on resistance and setup of thresholds (100 KΩ–1 GΩ|
|Relay||1 A at 30 V DC max.|
|Temperature range||21 °C to 30 °C for 1 GΩ setup shoes |
5 °C bis 30 °C for all other thresholds